#               << RESP_VER	RESP.XX.NR001..LHZ.130.1.1	2007.109	NEW	 >>
#		<< IRIS SEED Reader, Release 4.6 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     NR001
B050F16     Network:     XX
B052F03     Location:    ??
B052F04     Channel:     LHZ
B052F22     Start date:  2006,001,00:00:00.0000
B052F23     End date:    2999,365,23:59:59.0000
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros), NR001 ch LHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1
B053F08     Normalization frequency:               1
B053F09     Number of zeroes:                      0
B053F14     Number of poles:                       0
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
#		Complex poles:
#		  i  real          imag          real_error    imag_error
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  1
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  1.000000E+00  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.024000E+05
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  6.291300E+05
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 4
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  29
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  9.765620E-04  0.000000E+00
B054F08-09    2  2.441410E-03  0.000000E+00
B054F08-09    3  4.882810E-03  0.000000E+00
B054F08-09    4  8.544920E-03  0.000000E+00
B054F08-09    5  1.367190E-02  0.000000E+00
B054F08-09    6  2.050780E-02  0.000000E+00
B054F08-09    7  2.929690E-02  0.000000E+00
B054F08-09    8  3.930660E-02  0.000000E+00
B054F08-09    9  4.980470E-02  0.000000E+00
B054F08-09   10  6.005860E-02  0.000000E+00
B054F08-09   11  6.933590E-02  0.000000E+00
B054F08-09   12  7.690430E-02  0.000000E+00
B054F08-09   13  8.203120E-02  0.000000E+00
B054F08-09   14  8.398440E-02  0.000000E+00
B054F08-09   15  8.203120E-02  0.000000E+00
B054F08-09   16  7.690430E-02  0.000000E+00
B054F08-09   17  6.933590E-02  0.000000E+00
B054F08-09   18  6.005860E-02  0.000000E+00
B054F08-09   19  4.980470E-02  0.000000E+00
B054F08-09   20  3.930660E-02  0.000000E+00
B054F08-09   21  2.929690E-02  0.000000E+00
B054F08-09   22  2.050780E-02  0.000000E+00
B054F08-09   23  1.367190E-02  0.000000E+00
B054F08-09   24  8.544920E-03  0.000000E+00
B054F08-09   25  4.882810E-03  0.000000E+00
B054F08-09   26  2.441410E-03  0.000000E+00
B054F08-09   27  9.765620E-04  0.000000E+00
B054F08-09   28  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 4
B057F04     Input sample rate:                     1.024000E+05
B057F05     Decimation factor:                     8
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             1.367200E-04
B057F08     Correction applied (seconds):          1.367200E-04
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 4
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 5
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  13
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  2.929690E-03  0.000000E+00
B054F08-09    2  1.611330E-02  0.000000E+00
B054F08-09    3  5.371090E-02  0.000000E+00
B054F08-09    4  1.208500E-01  0.000000E+00
B054F08-09    5  1.933590E-01  0.000000E+00
B054F08-09    6  2.255860E-01  0.000000E+00
B054F08-09    7  1.933590E-01  0.000000E+00
B054F08-09    8  1.208500E-01  0.000000E+00
B054F08-09    9  5.371090E-02  0.000000E+00
B054F08-09   10  1.611330E-02  0.000000E+00
B054F08-09   11  2.929690E-03  0.000000E+00
B054F08-09   12  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 5
B057F04     Input sample rate:                     1.280000E+04
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             4.687500E-04
B057F08     Correction applied (seconds):          4.687500E-04
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 5
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 6
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  13
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  2.929690E-03  0.000000E+00
B054F08-09    2  1.611330E-02  0.000000E+00
B054F08-09    3  5.371090E-02  0.000000E+00
B054F08-09    4  1.208500E-01  0.000000E+00
B054F08-09    5  1.933590E-01  0.000000E+00
B054F08-09    6  2.255860E-01  0.000000E+00
B054F08-09    7  1.933590E-01  0.000000E+00
B054F08-09    8  1.208500E-01  0.000000E+00
B054F08-09    9  5.371090E-02  0.000000E+00
B054F08-09   10  1.611330E-02  0.000000E+00
B054F08-09   11  2.929690E-03  0.000000E+00
B054F08-09   12  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 6
B057F04     Input sample rate:                     6.400000E+03
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             9.375000E-04
B057F08     Correction applied (seconds):          9.375000E-04
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 6
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 7
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  13
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  2.929690E-03  0.000000E+00
B054F08-09    2  1.611330E-02  0.000000E+00
B054F08-09    3  5.371090E-02  0.000000E+00
B054F08-09    4  1.208500E-01  0.000000E+00
B054F08-09    5  1.933590E-01  0.000000E+00
B054F08-09    6  2.255860E-01  0.000000E+00
B054F08-09    7  1.933590E-01  0.000000E+00
B054F08-09    8  1.208500E-01  0.000000E+00
B054F08-09    9  5.371090E-02  0.000000E+00
B054F08-09   10  1.611330E-02  0.000000E+00
B054F08-09   11  2.929690E-03  0.000000E+00
B054F08-09   12  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 7
B057F04     Input sample rate:                     3.200000E+03
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             1.875000E-03
B057F08     Correction applied (seconds):          1.875000E-03
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 7
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 8
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  13
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  2.929690E-03  0.000000E+00
B054F08-09    2  1.611330E-02  0.000000E+00
B054F08-09    3  5.371090E-02  0.000000E+00
B054F08-09    4  1.208500E-01  0.000000E+00
B054F08-09    5  1.933590E-01  0.000000E+00
B054F08-09    6  2.255860E-01  0.000000E+00
B054F08-09    7  1.933590E-01  0.000000E+00
B054F08-09    8  1.208500E-01  0.000000E+00
B054F08-09    9  5.371090E-02  0.000000E+00
B054F08-09   10  1.611330E-02  0.000000E+00
B054F08-09   11  2.929690E-03  0.000000E+00
B054F08-09   12  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 8
B057F04     Input sample rate:                     1.600000E+03
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             3.750000E-03
B057F08     Correction applied (seconds):          3.750000E-03
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 8
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 9
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  13
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0  2.441410E-04  0.000000E+00
B054F08-09    1  2.929690E-03  0.000000E+00
B054F08-09    2  1.611330E-02  0.000000E+00
B054F08-09    3  5.371090E-02  0.000000E+00
B054F08-09    4  1.208500E-01  0.000000E+00
B054F08-09    5  1.933590E-01  0.000000E+00
B054F08-09    6  2.255860E-01  0.000000E+00
B054F08-09    7  1.933590E-01  0.000000E+00
B054F08-09    8  1.208500E-01  0.000000E+00
B054F08-09    9  5.371090E-02  0.000000E+00
B054F08-09   10  1.611330E-02  0.000000E+00
B054F08-09   11  2.929690E-03  0.000000E+00
B054F08-09   12  2.441410E-04  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 9
B057F04     Input sample rate:                     8.000000E+02
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             7.500000E-03
B057F08     Correction applied (seconds):          7.500000E-03
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 9
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 10
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  101
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -7.150320E-07  0.000000E+00
B054F08-09    1 -5.601090E-06  0.000000E+00
B054F08-09    2 -2.621790E-06  0.000000E+00
B054F08-09    3 -4.314030E-05  0.000000E+00
B054F08-09    4 -4.647710E-06  0.000000E+00
B054F08-09    5  1.430060E-06  0.000000E+00
B054F08-09    6  2.347690E-05  0.000000E+00
B054F08-09    7  1.430060E-06  0.000000E+00
B054F08-09    8 -5.279320E-05  0.000000E+00
B054F08-09    9 -3.666920E-04  0.000000E+00
B054F08-09   10  3.761070E-04  0.000000E+00
B054F08-09   11  8.542260E-04  0.000000E+00
B054F08-09   12  3.050810E-05  0.000000E+00
B054F08-09   13 -1.276210E-03  0.000000E+00
B054F08-09   14 -9.109510E-04  0.000000E+00
B054F08-09   15  1.276690E-03  0.000000E+00
B054F08-09   16  2.151650E-03  0.000000E+00
B054F08-09   17 -4.615540E-04  0.000000E+00
B054F08-09   18 -3.337650E-03  0.000000E+00
B054F08-09   19 -1.409330E-03  0.000000E+00
B054F08-09   20  3.770720E-03  0.000000E+00
B054F08-09   21  4.194140E-03  0.000000E+00
B054F08-09   22 -2.642880E-03  0.000000E+00
B054F08-09   23 -7.201210E-03  0.000000E+00
B054F08-09   24 -6.440060E-04  0.000000E+00
B054F08-09   25  9.184000E-03  0.000000E+00
B054F08-09   26  6.084450E-03  0.000000E+00
B054F08-09   27 -8.578240E-03  0.000000E+00
B054F08-09   28 -1.274010E-02  0.000000E+00
B054F08-09   29  3.982250E-03  0.000000E+00
B054F08-09   30  1.862610E-02  0.000000E+00
B054F08-09   31  5.205200E-03  0.000000E+00
B054F08-09   32 -2.094070E-02  0.000000E+00
B054F08-09   33 -1.816290E-02  0.000000E+00
B054F08-09   34  1.666690E-02  0.000000E+00
B054F08-09   35  3.224470E-02  0.000000E+00
B054F08-09   36 -3.465880E-03  0.000000E+00
B054F08-09   37 -4.295280E-02  0.000000E+00
B054F08-09   38 -1.932650E-02  0.000000E+00
B054F08-09   39  4.430900E-02  0.000000E+00
B054F08-09   40  4.979090E-02  0.000000E+00
B054F08-09   41 -2.941640E-02  0.000000E+00
B054F08-09   42 -8.260780E-02  0.000000E+00
B054F08-09   43 -9.341660E-03  0.000000E+00
B054F08-09   44  1.075520E-01  0.000000E+00
B054F08-09   45  8.166040E-02  0.000000E+00
B054F08-09   46 -1.031100E-01  0.000000E+00
B054F08-09   47 -2.042080E-01  0.000000E+00
B054F08-09   48 -3.122310E-05  0.000000E+00
B054F08-09   49  3.904320E-01  0.000000E+00
B054F08-09   50  5.899580E-01  0.000000E+00
B054F08-09   51  3.904320E-01  0.000000E+00
B054F08-09   52 -3.122310E-05  0.000000E+00
B054F08-09   53 -2.042080E-01  0.000000E+00
B054F08-09   54 -1.031100E-01  0.000000E+00
B054F08-09   55  8.166040E-02  0.000000E+00
B054F08-09   56  1.075520E-01  0.000000E+00
B054F08-09   57 -9.341660E-03  0.000000E+00
B054F08-09   58 -8.260780E-02  0.000000E+00
B054F08-09   59 -2.941640E-02  0.000000E+00
B054F08-09   60  4.979090E-02  0.000000E+00
B054F08-09   61  4.430900E-02  0.000000E+00
B054F08-09   62 -1.932650E-02  0.000000E+00
B054F08-09   63 -4.295280E-02  0.000000E+00
B054F08-09   64 -3.465880E-03  0.000000E+00
B054F08-09   65  3.224470E-02  0.000000E+00
B054F08-09   66  1.666690E-02  0.000000E+00
B054F08-09   67 -1.816290E-02  0.000000E+00
B054F08-09   68 -2.094070E-02  0.000000E+00
B054F08-09   69  5.205200E-03  0.000000E+00
B054F08-09   70  1.862610E-02  0.000000E+00
B054F08-09   71  3.982250E-03  0.000000E+00
B054F08-09   72 -1.274010E-02  0.000000E+00
B054F08-09   73 -8.578240E-03  0.000000E+00
B054F08-09   74  6.084450E-03  0.000000E+00
B054F08-09   75  9.184000E-03  0.000000E+00
B054F08-09   76 -6.440060E-04  0.000000E+00
B054F08-09   77 -7.201210E-03  0.000000E+00
B054F08-09   78 -2.642880E-03  0.000000E+00
B054F08-09   79  4.194140E-03  0.000000E+00
B054F08-09   80  3.770720E-03  0.000000E+00
B054F08-09   81 -1.409330E-03  0.000000E+00
B054F08-09   82 -3.337650E-03  0.000000E+00
B054F08-09   83 -4.615540E-04  0.000000E+00
B054F08-09   84  2.151650E-03  0.000000E+00
B054F08-09   85  1.276690E-03  0.000000E+00
B054F08-09   86 -9.109510E-04  0.000000E+00
B054F08-09   87 -1.276210E-03  0.000000E+00
B054F08-09   88  3.050810E-05  0.000000E+00
B054F08-09   89  8.542260E-04  0.000000E+00
B054F08-09   90  3.761070E-04  0.000000E+00
B054F08-09   91 -3.666920E-04  0.000000E+00
B054F08-09   92 -4.103100E-04  0.000000E+00
B054F08-09   93  2.526450E-05  0.000000E+00
B054F08-09   94  2.618210E-04  0.000000E+00
B054F08-09   95  1.206020E-04  0.000000E+00
B054F08-09   96 -9.998540E-05  0.000000E+00
B054F08-09   97 -1.623120E-04  0.000000E+00
B054F08-09   98 -9.795950E-05  0.000000E+00
B054F08-09   99 -2.943550E-05  0.000000E+00
B054F08-09  100 -3.098470E-06  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 10
B057F04     Input sample rate:                     4.000000E+02
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             1.250000E-01
B057F08     Correction applied (seconds):          1.250000E-01
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 10
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 11
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  235
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -1.098890E-05  0.000000E+00
B054F08-09    1 -1.997980E-05  0.000000E+00
B054F08-09    2 -3.296680E-05  0.000000E+00
B054F08-09    3 -4.395610E-05  0.000000E+00
B054F08-09    4 -4.795220E-05  0.000000E+00
B054F08-09    5 -4.095890E-05  0.000000E+00
B054F08-09    6 -1.898100E-05  0.000000E+00
B054F08-09    7  1.898100E-05  0.000000E+00
B054F08-09    8  6.793200E-05  0.000000E+00
B054F08-09    9  1.188810E-04  0.000000E+00
B054F08-09   10  1.588420E-04  0.000000E+00
B054F08-09   11  1.748260E-04  0.000000E+00
B054F08-09   12  1.578430E-04  0.000000E+00
B054F08-09   13  1.048950E-04  0.000000E+00
B054F08-09   14  2.497510E-05  0.000000E+00
B054F08-09   15 -6.493520E-05  0.000000E+00
B054F08-09   16 -1.408600E-04  0.000000E+00
B054F08-09   17 -1.788220E-04  0.000000E+00
B054F08-09   18 -1.608400E-04  0.000000E+00
B054F08-09   19 -8.591420E-05  0.000000E+00
B054F08-09   20  3.296680E-05  0.000000E+00
B054F08-09   21  1.638370E-04  0.000000E+00
B054F08-09   22  2.687330E-04  0.000000E+00
B054F08-09   23  3.106910E-04  0.000000E+00
B054F08-09   24  2.637370E-04  0.000000E+00
B054F08-09   25  1.308700E-04  0.000000E+00
B054F08-09   26 -6.093910E-05  0.000000E+00
B054F08-09   27 -2.607400E-04  0.000000E+00
B054F08-09   28 -4.085930E-04  0.000000E+00
B054F08-09   29 -4.485540E-04  0.000000E+00
B054F08-09   30 -3.536480E-04  0.000000E+00
B054F08-09   31 -1.358640E-04  0.000000E+00
B054F08-09   32  1.558450E-04  0.000000E+00
B054F08-09   33  4.385630E-04  0.000000E+00
B054F08-09   34  6.233790E-04  0.000000E+00
B054F08-09   35  6.383650E-04  0.000000E+00
B054F08-09   36  4.565460E-04  0.000000E+00
B054F08-09   37  1.088910E-04  0.000000E+00
B054F08-09   38 -3.156860E-04  0.000000E+00
B054F08-09   39 -6.943090E-04  0.000000E+00
B054F08-09   40 -9.031010E-04  0.000000E+00
B054F08-09   41 -8.541500E-04  0.000000E+00
B054F08-09   42 -5.334690E-04  0.000000E+00
B054F08-09   43 -7.991640E-06  0.000000E+00
B054F08-09   44  5.814210E-04  0.000000E+00
B054F08-09   45  1.056950E-03  0.000000E+00
B054F08-09   46  1.256750E-03  0.000000E+00
B054F08-09   47  1.087920E-03  0.000000E+00
B054F08-09   48  5.594430E-04  0.000000E+00
B054F08-09   49 -2.017990E-04  0.000000E+00
B054F08-09   50 -9.830210E-04  0.000000E+00
B054F08-09   51 -1.540470E-03  0.000000E+00
B054F08-09   52 -1.677330E-03  0.000000E+00
B054F08-09   53 -1.303700E-03  0.000000E+00
B054F08-09   54 -4.845180E-04  0.000000E+00
B054F08-09   55  5.714310E-04  0.000000E+00
B054F08-09   56  1.556450E-03  0.000000E+00
B054F08-09   57  2.156850E-03  0.000000E+00
B054F08-09   58  2.142870E-03  0.000000E+00
B054F08-09   59  1.458550E-03  0.000000E+00
B054F08-09   60  2.507500E-04  0.000000E+00
B054F08-09   61 -1.153850E-03  0.000000E+00
B054F08-09   62 -2.335680E-03  0.000000E+00
B054F08-09   63 -2.903110E-03  0.000000E+00
B054F08-09   64 -2.617400E-03  0.000000E+00
B054F08-09   65 -1.487520E-03  0.000000E+00
B054F08-09   66  2.157850E-04  0.000000E+00
B054F08-09   67  2.014000E-03  0.000000E+00
B054F08-09   68  3.351660E-03  0.000000E+00
B054F08-09   69  3.768250E-03  0.000000E+00
B054F08-09   70  3.045970E-03  0.000000E+00
B054F08-09   71  1.303700E-03  0.000000E+00
B054F08-09   72 -1.009000E-03  0.000000E+00
B054F08-09   73 -3.220800E-03  0.000000E+00
B054F08-09   74 -4.631390E-03  0.000000E+00
B054F08-09   75 -4.723300E-03  0.000000E+00
B054F08-09   76 -3.346670E-03  0.000000E+00
B054F08-09   77 -7.932110E-04  0.000000E+00
B054F08-09   78  2.244770E-03  0.000000E+00
B054F08-09   79  4.865160E-03  0.000000E+00
B054F08-09   80  6.205830E-03  0.000000E+00
B054F08-09   81  5.727300E-03  0.000000E+00
B054F08-09   82  3.408610E-03  0.000000E+00
B054F08-09   83 -1.998010E-04  0.000000E+00
B054F08-09   84 -4.091930E-03  0.000000E+00
B054F08-09   85 -7.075960E-03  0.000000E+00
B054F08-09   86 -8.127910E-03  0.000000E+00
B054F08-09   87 -6.728310E-03  0.000000E+00
B054F08-09   88 -3.071940E-03  0.000000E+00
B054F08-09   89  1.923090E-03  0.000000E+00
B054F08-09   90  6.827210E-03  0.000000E+00
B054F08-09   91  1.009100E-02  0.000000E+00
B054F08-09   92  1.051750E-02  0.000000E+00
B054F08-09   93  7.664370E-03  0.000000E+00
B054F08-09   94  2.065940E-03  0.000000E+00
B054F08-09   95 -4.832190E-03  0.000000E+00
B054F08-09   96 -1.101000E-02  0.000000E+00
B054F08-09   97 -1.443760E-02  0.000000E+00
B054F08-09   98 -1.369340E-02  0.000000E+00
B054F08-09   99 -8.474570E-03  0.000000E+00
B054F08-09  100  1.738270E-04  0.000000E+00
B054F08-09  101  1.000400E-02  0.000000E+00
B054F08-09  102  1.808500E-02  0.000000E+00
B054F08-09  103  2.159350E-02  0.000000E+00
B054F08-09  104  1.866640E-02  0.000000E+00
B054F08-09  105  9.100940E-03  0.000000E+00
B054F08-09  106 -5.328700E-03  0.000000E+00
B054F08-09  107 -2.105410E-02  0.000000E+00
B054F08-09  108 -3.339580E-02  0.000000E+00
B054F08-09  109 -3.762260E-02  0.000000E+00
B054F08-09  110 -3.013700E-02  0.000000E+00
B054F08-09  111 -9.497550E-03  0.000000E+00
B054F08-09  112  2.299310E-02  0.000000E+00
B054F08-09  113  6.330400E-02  0.000000E+00
B054F08-09  114  1.053400E-01  0.000000E+00
B054F08-09  115  1.421240E-01  0.000000E+00
B054F08-09  116  1.672260E-01  0.000000E+00
B054F08-09  117  1.761340E-01  0.000000E+00
B054F08-09  118  1.672260E-01  0.000000E+00
B054F08-09  119  1.421240E-01  0.000000E+00
B054F08-09  120  1.053400E-01  0.000000E+00
B054F08-09  121  6.330400E-02  0.000000E+00
B054F08-09  122  2.299310E-02  0.000000E+00
B054F08-09  123 -9.497550E-03  0.000000E+00
B054F08-09  124 -3.013700E-02  0.000000E+00
B054F08-09  125 -3.762260E-02  0.000000E+00
B054F08-09  126 -3.339580E-02  0.000000E+00
B054F08-09  127 -2.105410E-02  0.000000E+00
B054F08-09  128 -5.328700E-03  0.000000E+00
B054F08-09  129  9.100940E-03  0.000000E+00
B054F08-09  130  1.866640E-02  0.000000E+00
B054F08-09  131  2.159350E-02  0.000000E+00
B054F08-09  132  1.808500E-02  0.000000E+00
B054F08-09  133  1.000400E-02  0.000000E+00
B054F08-09  134  1.738270E-04  0.000000E+00
B054F08-09  135 -8.474570E-03  0.000000E+00
B054F08-09  136 -1.369340E-02  0.000000E+00
B054F08-09  137 -1.443760E-02  0.000000E+00
B054F08-09  138 -1.101000E-02  0.000000E+00
B054F08-09  139 -4.832190E-03  0.000000E+00
B054F08-09  140  2.065940E-03  0.000000E+00
B054F08-09  141  7.664370E-03  0.000000E+00
B054F08-09  142  1.051750E-02  0.000000E+00
B054F08-09  143  1.009100E-02  0.000000E+00
B054F08-09  144  6.827210E-03  0.000000E+00
B054F08-09  145  1.923090E-03  0.000000E+00
B054F08-09  146 -3.071940E-03  0.000000E+00
B054F08-09  147 -6.728310E-03  0.000000E+00
B054F08-09  148 -8.127910E-03  0.000000E+00
B054F08-09  149 -7.075960E-03  0.000000E+00
B054F08-09  150 -4.091930E-03  0.000000E+00
B054F08-09  151 -1.998010E-04  0.000000E+00
B054F08-09  152  3.408610E-03  0.000000E+00
B054F08-09  153  5.727300E-03  0.000000E+00
B054F08-09  154  6.205830E-03  0.000000E+00
B054F08-09  155  4.865160E-03  0.000000E+00
B054F08-09  156  2.244770E-03  0.000000E+00
B054F08-09  157 -7.932110E-04  0.000000E+00
B054F08-09  158 -3.346670E-03  0.000000E+00
B054F08-09  159 -4.723300E-03  0.000000E+00
B054F08-09  160 -4.631390E-03  0.000000E+00
B054F08-09  161 -3.220800E-03  0.000000E+00
B054F08-09  162 -1.009000E-03  0.000000E+00
B054F08-09  163  1.303700E-03  0.000000E+00
B054F08-09  164  3.045970E-03  0.000000E+00
B054F08-09  165  3.768250E-03  0.000000E+00
B054F08-09  166  3.351660E-03  0.000000E+00
B054F08-09  167  2.014000E-03  0.000000E+00
B054F08-09  168  2.157850E-04  0.000000E+00
B054F08-09  169 -1.487520E-03  0.000000E+00
B054F08-09  170 -2.617400E-03  0.000000E+00
B054F08-09  171 -2.903110E-03  0.000000E+00
B054F08-09  172 -2.335680E-03  0.000000E+00
B054F08-09  173 -1.153850E-03  0.000000E+00
B054F08-09  174  2.507500E-04  0.000000E+00
B054F08-09  175  1.458550E-03  0.000000E+00
B054F08-09  176  2.142870E-03  0.000000E+00
B054F08-09  177  2.156850E-03  0.000000E+00
B054F08-09  178  1.556450E-03  0.000000E+00
B054F08-09  179  5.714310E-04  0.000000E+00
B054F08-09  180 -4.845180E-04  0.000000E+00
B054F08-09  181 -1.303700E-03  0.000000E+00
B054F08-09  182 -1.677330E-03  0.000000E+00
B054F08-09  183 -1.540470E-03  0.000000E+00
B054F08-09  184 -9.830210E-04  0.000000E+00
B054F08-09  185 -2.017990E-04  0.000000E+00
B054F08-09  186  5.594430E-04  0.000000E+00
B054F08-09  187  1.087920E-03  0.000000E+00
B054F08-09  188  1.256750E-03  0.000000E+00
B054F08-09  189  1.056950E-03  0.000000E+00
B054F08-09  190  5.814210E-04  0.000000E+00
B054F08-09  191 -7.991640E-06  0.000000E+00
B054F08-09  192 -5.334690E-04  0.000000E+00
B054F08-09  193 -8.541500E-04  0.000000E+00
B054F08-09  194 -9.031010E-04  0.000000E+00
B054F08-09  195 -6.943090E-04  0.000000E+00
B054F08-09  196 -3.156860E-04  0.000000E+00
B054F08-09  197  1.088910E-04  0.000000E+00
B054F08-09  198  4.565460E-04  0.000000E+00
B054F08-09  199  6.383650E-04  0.000000E+00
B054F08-09  200  6.233790E-04  0.000000E+00
B054F08-09  201  4.385630E-04  0.000000E+00
B054F08-09  202  1.558450E-04  0.000000E+00
B054F08-09  203 -1.358640E-04  0.000000E+00
B054F08-09  204 -3.536480E-04  0.000000E+00
B054F08-09  205 -4.485540E-04  0.000000E+00
B054F08-09  206 -4.085930E-04  0.000000E+00
B054F08-09  207 -2.607400E-04  0.000000E+00
B054F08-09  208 -6.093910E-05  0.000000E+00
B054F08-09  209  1.308700E-04  0.000000E+00
B054F08-09  210  2.637370E-04  0.000000E+00
B054F08-09  211  3.106910E-04  0.000000E+00
B054F08-09  212  2.687330E-04  0.000000E+00
B054F08-09  213  1.638370E-04  0.000000E+00
B054F08-09  214  3.296680E-05  0.000000E+00
B054F08-09  215 -8.591420E-05  0.000000E+00
B054F08-09  216 -1.608400E-04  0.000000E+00
B054F08-09  217 -1.788220E-04  0.000000E+00
B054F08-09  218 -1.408600E-04  0.000000E+00
B054F08-09  219 -6.493520E-05  0.000000E+00
B054F08-09  220  2.497510E-05  0.000000E+00
B054F08-09  221  1.048950E-04  0.000000E+00
B054F08-09  222  1.578430E-04  0.000000E+00
B054F08-09  223  1.748260E-04  0.000000E+00
B054F08-09  224  1.588420E-04  0.000000E+00
B054F08-09  225  1.188810E-04  0.000000E+00
B054F08-09  226  6.793200E-05  0.000000E+00
B054F08-09  227  1.898100E-05  0.000000E+00
B054F08-09  228 -1.898100E-05  0.000000E+00
B054F08-09  229 -4.095890E-05  0.000000E+00
B054F08-09  230 -4.795220E-05  0.000000E+00
B054F08-09  231 -4.395610E-05  0.000000E+00
B054F08-09  232 -3.296680E-05  0.000000E+00
B054F08-09  233 -1.997980E-05  0.000000E+00
B054F08-09  234 -1.098890E-05  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 11
B057F04     Input sample rate:                     2.000000E+02
B057F05     Decimation factor:                     5
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             5.850000E-01
B057F08     Correction applied (seconds):          5.850000E-01
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 11
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 12
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  95
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -4.704200E-05  0.000000E+00
B054F08-09    1 -1.861670E-04  0.000000E+00
B054F08-09    2 -2.922630E-04  0.000000E+00
B054F08-09    3 -1.451310E-04  0.000000E+00
B054F08-09    4  2.121910E-04  0.000000E+00
B054F08-09    5  3.102790E-04  0.000000E+00
B054F08-09    6 -1.241110E-04  0.000000E+00
B054F08-09    7 -5.234710E-04  0.000000E+00
B054F08-09    8 -1.141030E-04  0.000000E+00
B054F08-09    9  6.926240E-04  0.000000E+00
B054F08-09   10  5.344810E-04  0.000000E+00
B054F08-09   11 -6.956270E-04  0.000000E+00
B054F08-09   12 -1.099990E-03  0.000000E+00
B054F08-09   13  4.063660E-04  0.000000E+00
B054F08-09   14  1.698530E-03  0.000000E+00
B054F08-09   15  2.732460E-04  0.000000E+00
B054F08-09   16 -2.139930E-03  0.000000E+00
B054F08-09   17 -1.365230E-03  0.000000E+00
B054F08-09   18  2.180960E-03  0.000000E+00
B054F08-09   19  2.778500E-03  0.000000E+00
B054F08-09   20 -1.568410E-03  0.000000E+00
B054F08-09   21 -4.278850E-03  0.000000E+00
B054F08-09   22  1.080970E-04  0.000000E+00
B054F08-09   23  5.494950E-03  0.000000E+00
B054F08-09   24  2.263040E-03  0.000000E+00
B054F08-09   25 -5.950360E-03  0.000000E+00
B054F08-09   26 -5.410870E-03  0.000000E+00
B054F08-09   27  5.135630E-03  0.000000E+00
B054F08-09   28  8.955070E-03  0.000000E+00
B054F08-09   29 -2.601340E-03  0.000000E+00
B054F08-09   30 -1.226510E-02  0.000000E+00
B054F08-09   31 -1.939750E-03  0.000000E+00
B054F08-09   32  1.447800E-02  0.000000E+00
B054F08-09   33  8.519680E-03  0.000000E+00
B054F08-09   34 -1.456010E-02  0.000000E+00
B054F08-09   35 -1.686520E-02  0.000000E+00
B054F08-09   36  1.134020E-02  0.000000E+00
B054F08-09   37  2.638380E-02  0.000000E+00
B054F08-09   38 -3.446100E-03  0.000000E+00
B054F08-09   39 -3.622360E-02  0.000000E+00
B054F08-09   40 -1.110300E-02  0.000000E+00
B054F08-09   41  4.537790E-02  0.000000E+00
B054F08-09   42  3.664000E-02  0.000000E+00
B054F08-09   43 -5.282360E-02  0.000000E+00
B054F08-09   44 -8.887210E-02  0.000000E+00
B054F08-09   45  5.768800E-02  0.000000E+00
B054F08-09   46  3.126340E-01  0.000000E+00
B054F08-09   47  4.410720E-01  0.000000E+00
B054F08-09   48  3.126340E-01  0.000000E+00
B054F08-09   49  5.768800E-02  0.000000E+00
B054F08-09   50 -8.887210E-02  0.000000E+00
B054F08-09   51 -5.282360E-02  0.000000E+00
B054F08-09   52  3.664000E-02  0.000000E+00
B054F08-09   53  4.537790E-02  0.000000E+00
B054F08-09   54 -1.110300E-02  0.000000E+00
B054F08-09   55 -3.622360E-02  0.000000E+00
B054F08-09   56 -3.446100E-03  0.000000E+00
B054F08-09   57  2.638380E-02  0.000000E+00
B054F08-09   58  1.134020E-02  0.000000E+00
B054F08-09   59 -1.686520E-02  0.000000E+00
B054F08-09   60 -1.456010E-02  0.000000E+00
B054F08-09   61  8.519680E-03  0.000000E+00
B054F08-09   62  1.447800E-02  0.000000E+00
B054F08-09   63 -1.939750E-03  0.000000E+00
B054F08-09   64 -1.226510E-02  0.000000E+00
B054F08-09   65 -2.601340E-03  0.000000E+00
B054F08-09   66  8.955070E-03  0.000000E+00
B054F08-09   67  5.135630E-03  0.000000E+00
B054F08-09   68 -5.410870E-03  0.000000E+00
B054F08-09   69 -5.950360E-03  0.000000E+00
B054F08-09   70  2.263040E-03  0.000000E+00
B054F08-09   71  5.494950E-03  0.000000E+00
B054F08-09   72  1.080970E-04  0.000000E+00
B054F08-09   73 -4.278850E-03  0.000000E+00
B054F08-09   74 -1.568410E-03  0.000000E+00
B054F08-09   75  2.778500E-03  0.000000E+00
B054F08-09   76  2.180960E-03  0.000000E+00
B054F08-09   77 -1.365230E-03  0.000000E+00
B054F08-09   78 -2.139930E-03  0.000000E+00
B054F08-09   79  2.732460E-04  0.000000E+00
B054F08-09   80  1.698530E-03  0.000000E+00
B054F08-09   81  4.063660E-04  0.000000E+00
B054F08-09   82 -1.099990E-03  0.000000E+00
B054F08-09   83 -6.956270E-04  0.000000E+00
B054F08-09   84  5.344810E-04  0.000000E+00
B054F08-09   85  6.926240E-04  0.000000E+00
B054F08-09   86 -1.141030E-04  0.000000E+00
B054F08-09   87 -5.234710E-04  0.000000E+00
B054F08-09   88 -1.241110E-04  0.000000E+00
B054F08-09   89  3.102790E-04  0.000000E+00
B054F08-09   90  2.121910E-04  0.000000E+00
B054F08-09   91 -1.451310E-04  0.000000E+00
B054F08-09   92 -2.922630E-04  0.000000E+00
B054F08-09   93 -1.861670E-04  0.000000E+00
B054F08-09   94 -4.704200E-05  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 12
B057F04     Input sample rate:                     4.000000E+01
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             1.175000E+00
B057F08     Correction applied (seconds):          1.175000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 12
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 13
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  95
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -4.704200E-05  0.000000E+00
B054F08-09    1 -1.861670E-04  0.000000E+00
B054F08-09    2 -2.922630E-04  0.000000E+00
B054F08-09    3 -1.451310E-04  0.000000E+00
B054F08-09    4  2.121910E-04  0.000000E+00
B054F08-09    5  3.102790E-04  0.000000E+00
B054F08-09    6 -1.241110E-04  0.000000E+00
B054F08-09    7 -5.234710E-04  0.000000E+00
B054F08-09    8 -1.141030E-04  0.000000E+00
B054F08-09    9  6.926240E-04  0.000000E+00
B054F08-09   10  5.344810E-04  0.000000E+00
B054F08-09   11 -6.956270E-04  0.000000E+00
B054F08-09   12 -1.099990E-03  0.000000E+00
B054F08-09   13  4.063660E-04  0.000000E+00
B054F08-09   14  1.698530E-03  0.000000E+00
B054F08-09   15  2.732460E-04  0.000000E+00
B054F08-09   16 -2.139930E-03  0.000000E+00
B054F08-09   17 -1.365230E-03  0.000000E+00
B054F08-09   18  2.180960E-03  0.000000E+00
B054F08-09   19  2.778500E-03  0.000000E+00
B054F08-09   20 -1.568410E-03  0.000000E+00
B054F08-09   21 -4.278850E-03  0.000000E+00
B054F08-09   22  1.080970E-04  0.000000E+00
B054F08-09   23  5.494950E-03  0.000000E+00
B054F08-09   24  2.263040E-03  0.000000E+00
B054F08-09   25 -5.950360E-03  0.000000E+00
B054F08-09   26 -5.410870E-03  0.000000E+00
B054F08-09   27  5.135630E-03  0.000000E+00
B054F08-09   28  8.955070E-03  0.000000E+00
B054F08-09   29 -2.601340E-03  0.000000E+00
B054F08-09   30 -1.226510E-02  0.000000E+00
B054F08-09   31 -1.939750E-03  0.000000E+00
B054F08-09   32  1.447800E-02  0.000000E+00
B054F08-09   33  8.519680E-03  0.000000E+00
B054F08-09   34 -1.456010E-02  0.000000E+00
B054F08-09   35 -1.686520E-02  0.000000E+00
B054F08-09   36  1.134020E-02  0.000000E+00
B054F08-09   37  2.638380E-02  0.000000E+00
B054F08-09   38 -3.446100E-03  0.000000E+00
B054F08-09   39 -3.622360E-02  0.000000E+00
B054F08-09   40 -1.110300E-02  0.000000E+00
B054F08-09   41  4.537790E-02  0.000000E+00
B054F08-09   42  3.664000E-02  0.000000E+00
B054F08-09   43 -5.282360E-02  0.000000E+00
B054F08-09   44 -8.887210E-02  0.000000E+00
B054F08-09   45  5.768800E-02  0.000000E+00
B054F08-09   46  3.126340E-01  0.000000E+00
B054F08-09   47  4.410720E-01  0.000000E+00
B054F08-09   48  3.126340E-01  0.000000E+00
B054F08-09   49  5.768800E-02  0.000000E+00
B054F08-09   50 -8.887210E-02  0.000000E+00
B054F08-09   51 -5.282360E-02  0.000000E+00
B054F08-09   52  3.664000E-02  0.000000E+00
B054F08-09   53  4.537790E-02  0.000000E+00
B054F08-09   54 -1.110300E-02  0.000000E+00
B054F08-09   55 -3.622360E-02  0.000000E+00
B054F08-09   56 -3.446100E-03  0.000000E+00
B054F08-09   57  2.638380E-02  0.000000E+00
B054F08-09   58  1.134020E-02  0.000000E+00
B054F08-09   59 -1.686520E-02  0.000000E+00
B054F08-09   60 -1.456010E-02  0.000000E+00
B054F08-09   61  8.519680E-03  0.000000E+00
B054F08-09   62  1.447800E-02  0.000000E+00
B054F08-09   63 -1.939750E-03  0.000000E+00
B054F08-09   64 -1.226510E-02  0.000000E+00
B054F08-09   65 -2.601340E-03  0.000000E+00
B054F08-09   66  8.955070E-03  0.000000E+00
B054F08-09   67  5.135630E-03  0.000000E+00
B054F08-09   68 -5.410870E-03  0.000000E+00
B054F08-09   69 -5.950360E-03  0.000000E+00
B054F08-09   70  2.263040E-03  0.000000E+00
B054F08-09   71  5.494950E-03  0.000000E+00
B054F08-09   72  1.080970E-04  0.000000E+00
B054F08-09   73 -4.278850E-03  0.000000E+00
B054F08-09   74 -1.568410E-03  0.000000E+00
B054F08-09   75  2.778500E-03  0.000000E+00
B054F08-09   76  2.180960E-03  0.000000E+00
B054F08-09   77 -1.365230E-03  0.000000E+00
B054F08-09   78 -2.139930E-03  0.000000E+00
B054F08-09   79  2.732460E-04  0.000000E+00
B054F08-09   80  1.698530E-03  0.000000E+00
B054F08-09   81  4.063660E-04  0.000000E+00
B054F08-09   82 -1.099990E-03  0.000000E+00
B054F08-09   83 -6.956270E-04  0.000000E+00
B054F08-09   84  5.344810E-04  0.000000E+00
B054F08-09   85  6.926240E-04  0.000000E+00
B054F08-09   86 -1.141030E-04  0.000000E+00
B054F08-09   87 -5.234710E-04  0.000000E+00
B054F08-09   88 -1.241110E-04  0.000000E+00
B054F08-09   89  3.102790E-04  0.000000E+00
B054F08-09   90  2.121910E-04  0.000000E+00
B054F08-09   91 -1.451310E-04  0.000000E+00
B054F08-09   92 -2.922630E-04  0.000000E+00
B054F08-09   93 -1.861670E-04  0.000000E+00
B054F08-09   94 -4.704200E-05  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 13
B057F04     Input sample rate:                     2.000000E+01
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             2.350000E+00
B057F08     Correction applied (seconds):          2.350000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 13
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 14
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  95
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -4.704200E-05  0.000000E+00
B054F08-09    1 -1.861670E-04  0.000000E+00
B054F08-09    2 -2.922630E-04  0.000000E+00
B054F08-09    3 -1.451310E-04  0.000000E+00
B054F08-09    4  2.121910E-04  0.000000E+00
B054F08-09    5  3.102790E-04  0.000000E+00
B054F08-09    6 -1.241110E-04  0.000000E+00
B054F08-09    7 -5.234710E-04  0.000000E+00
B054F08-09    8 -1.141030E-04  0.000000E+00
B054F08-09    9  6.926240E-04  0.000000E+00
B054F08-09   10  5.344810E-04  0.000000E+00
B054F08-09   11 -6.956270E-04  0.000000E+00
B054F08-09   12 -1.099990E-03  0.000000E+00
B054F08-09   13  4.063660E-04  0.000000E+00
B054F08-09   14  1.698530E-03  0.000000E+00
B054F08-09   15  2.732460E-04  0.000000E+00
B054F08-09   16 -2.139930E-03  0.000000E+00
B054F08-09   17 -1.365230E-03  0.000000E+00
B054F08-09   18  2.180960E-03  0.000000E+00
B054F08-09   19  2.778500E-03  0.000000E+00
B054F08-09   20 -1.568410E-03  0.000000E+00
B054F08-09   21 -4.278850E-03  0.000000E+00
B054F08-09   22  1.080970E-04  0.000000E+00
B054F08-09   23  5.494950E-03  0.000000E+00
B054F08-09   24  2.263040E-03  0.000000E+00
B054F08-09   25 -5.950360E-03  0.000000E+00
B054F08-09   26 -5.410870E-03  0.000000E+00
B054F08-09   27  5.135630E-03  0.000000E+00
B054F08-09   28  8.955070E-03  0.000000E+00
B054F08-09   29 -2.601340E-03  0.000000E+00
B054F08-09   30 -1.226510E-02  0.000000E+00
B054F08-09   31 -1.939750E-03  0.000000E+00
B054F08-09   32  1.447800E-02  0.000000E+00
B054F08-09   33  8.519680E-03  0.000000E+00
B054F08-09   34 -1.456010E-02  0.000000E+00
B054F08-09   35 -1.686520E-02  0.000000E+00
B054F08-09   36  1.134020E-02  0.000000E+00
B054F08-09   37  2.638380E-02  0.000000E+00
B054F08-09   38 -3.446100E-03  0.000000E+00
B054F08-09   39 -3.622360E-02  0.000000E+00
B054F08-09   40 -1.110300E-02  0.000000E+00
B054F08-09   41  4.537790E-02  0.000000E+00
B054F08-09   42  3.664000E-02  0.000000E+00
B054F08-09   43 -5.282360E-02  0.000000E+00
B054F08-09   44 -8.887210E-02  0.000000E+00
B054F08-09   45  5.768800E-02  0.000000E+00
B054F08-09   46  3.126340E-01  0.000000E+00
B054F08-09   47  4.410720E-01  0.000000E+00
B054F08-09   48  3.126340E-01  0.000000E+00
B054F08-09   49  5.768800E-02  0.000000E+00
B054F08-09   50 -8.887210E-02  0.000000E+00
B054F08-09   51 -5.282360E-02  0.000000E+00
B054F08-09   52  3.664000E-02  0.000000E+00
B054F08-09   53  4.537790E-02  0.000000E+00
B054F08-09   54 -1.110300E-02  0.000000E+00
B054F08-09   55 -3.622360E-02  0.000000E+00
B054F08-09   56 -3.446100E-03  0.000000E+00
B054F08-09   57  2.638380E-02  0.000000E+00
B054F08-09   58  1.134020E-02  0.000000E+00
B054F08-09   59 -1.686520E-02  0.000000E+00
B054F08-09   60 -1.456010E-02  0.000000E+00
B054F08-09   61  8.519680E-03  0.000000E+00
B054F08-09   62  1.447800E-02  0.000000E+00
B054F08-09   63 -1.939750E-03  0.000000E+00
B054F08-09   64 -1.226510E-02  0.000000E+00
B054F08-09   65 -2.601340E-03  0.000000E+00
B054F08-09   66  8.955070E-03  0.000000E+00
B054F08-09   67  5.135630E-03  0.000000E+00
B054F08-09   68 -5.410870E-03  0.000000E+00
B054F08-09   69 -5.950360E-03  0.000000E+00
B054F08-09   70  2.263040E-03  0.000000E+00
B054F08-09   71  5.494950E-03  0.000000E+00
B054F08-09   72  1.080970E-04  0.000000E+00
B054F08-09   73 -4.278850E-03  0.000000E+00
B054F08-09   74 -1.568410E-03  0.000000E+00
B054F08-09   75  2.778500E-03  0.000000E+00
B054F08-09   76  2.180960E-03  0.000000E+00
B054F08-09   77 -1.365230E-03  0.000000E+00
B054F08-09   78 -2.139930E-03  0.000000E+00
B054F08-09   79  2.732460E-04  0.000000E+00
B054F08-09   80  1.698530E-03  0.000000E+00
B054F08-09   81  4.063660E-04  0.000000E+00
B054F08-09   82 -1.099990E-03  0.000000E+00
B054F08-09   83 -6.956270E-04  0.000000E+00
B054F08-09   84  5.344810E-04  0.000000E+00
B054F08-09   85  6.926240E-04  0.000000E+00
B054F08-09   86 -1.141030E-04  0.000000E+00
B054F08-09   87 -5.234710E-04  0.000000E+00
B054F08-09   88 -1.241110E-04  0.000000E+00
B054F08-09   89  3.102790E-04  0.000000E+00
B054F08-09   90  2.121910E-04  0.000000E+00
B054F08-09   91 -1.451310E-04  0.000000E+00
B054F08-09   92 -2.922630E-04  0.000000E+00
B054F08-09   93 -1.861670E-04  0.000000E+00
B054F08-09   94 -4.704200E-05  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 14
B057F04     Input sample rate:                     1.000000E+01
B057F05     Decimation factor:                     2
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             4.700000E+00
B057F08     Correction applied (seconds):          4.700000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 14
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients), NR001 ch LHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 15
B054F05     Response in units lookup:              COUNTS - Digital Counts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  235
B054F10     Number of denominators:                0
#		Numerator coefficients:
#		  i, coefficient,  error
B054F08-09    0 -1.098890E-05  0.000000E+00
B054F08-09    1 -1.997980E-05  0.000000E+00
B054F08-09    2 -3.296680E-05  0.000000E+00
B054F08-09    3 -4.395610E-05  0.000000E+00
B054F08-09    4 -4.795220E-05  0.000000E+00
B054F08-09    5 -4.095890E-05  0.000000E+00
B054F08-09    6 -1.898100E-05  0.000000E+00
B054F08-09    7  1.898100E-05  0.000000E+00
B054F08-09    8  6.793200E-05  0.000000E+00
B054F08-09    9  1.188810E-04  0.000000E+00
B054F08-09   10  1.588420E-04  0.000000E+00
B054F08-09   11  1.748260E-04  0.000000E+00
B054F08-09   12  1.578430E-04  0.000000E+00
B054F08-09   13  1.048950E-04  0.000000E+00
B054F08-09   14  2.497510E-05  0.000000E+00
B054F08-09   15 -6.493520E-05  0.000000E+00
B054F08-09   16 -1.408600E-04  0.000000E+00
B054F08-09   17 -1.788220E-04  0.000000E+00
B054F08-09   18 -1.608400E-04  0.000000E+00
B054F08-09   19 -8.591420E-05  0.000000E+00
B054F08-09   20  3.296680E-05  0.000000E+00
B054F08-09   21  1.638370E-04  0.000000E+00
B054F08-09   22  2.687330E-04  0.000000E+00
B054F08-09   23  3.106910E-04  0.000000E+00
B054F08-09   24  2.637370E-04  0.000000E+00
B054F08-09   25  1.308700E-04  0.000000E+00
B054F08-09   26 -6.093910E-05  0.000000E+00
B054F08-09   27 -2.607400E-04  0.000000E+00
B054F08-09   28 -4.085930E-04  0.000000E+00
B054F08-09   29 -4.485540E-04  0.000000E+00
B054F08-09   30 -3.536480E-04  0.000000E+00
B054F08-09   31 -1.358640E-04  0.000000E+00
B054F08-09   32  1.558450E-04  0.000000E+00
B054F08-09   33  4.385630E-04  0.000000E+00
B054F08-09   34  6.233790E-04  0.000000E+00
B054F08-09   35  6.383650E-04  0.000000E+00
B054F08-09   36  4.565460E-04  0.000000E+00
B054F08-09   37  1.088910E-04  0.000000E+00
B054F08-09   38 -3.156860E-04  0.000000E+00
B054F08-09   39 -6.943090E-04  0.000000E+00
B054F08-09   40 -9.031010E-04  0.000000E+00
B054F08-09   41 -8.541500E-04  0.000000E+00
B054F08-09   42 -5.334690E-04  0.000000E+00
B054F08-09   43 -7.991640E-06  0.000000E+00
B054F08-09   44  5.814210E-04  0.000000E+00
B054F08-09   45  1.056950E-03  0.000000E+00
B054F08-09   46  1.256750E-03  0.000000E+00
B054F08-09   47  1.087920E-03  0.000000E+00
B054F08-09   48  5.594430E-04  0.000000E+00
B054F08-09   49 -2.017990E-04  0.000000E+00
B054F08-09   50 -9.830210E-04  0.000000E+00
B054F08-09   51 -1.540470E-03  0.000000E+00
B054F08-09   52 -1.677330E-03  0.000000E+00
B054F08-09   53 -1.303700E-03  0.000000E+00
B054F08-09   54 -4.845180E-04  0.000000E+00
B054F08-09   55  5.714310E-04  0.000000E+00
B054F08-09   56  1.556450E-03  0.000000E+00
B054F08-09   57  2.156850E-03  0.000000E+00
B054F08-09   58  2.142870E-03  0.000000E+00
B054F08-09   59  1.458550E-03  0.000000E+00
B054F08-09   60  2.507500E-04  0.000000E+00
B054F08-09   61 -1.153850E-03  0.000000E+00
B054F08-09   62 -2.335680E-03  0.000000E+00
B054F08-09   63 -2.903110E-03  0.000000E+00
B054F08-09   64 -2.617400E-03  0.000000E+00
B054F08-09   65 -1.487520E-03  0.000000E+00
B054F08-09   66  2.157850E-04  0.000000E+00
B054F08-09   67  2.014000E-03  0.000000E+00
B054F08-09   68  3.351660E-03  0.000000E+00
B054F08-09   69  3.768250E-03  0.000000E+00
B054F08-09   70  3.045970E-03  0.000000E+00
B054F08-09   71  1.303700E-03  0.000000E+00
B054F08-09   72 -1.009000E-03  0.000000E+00
B054F08-09   73 -3.220800E-03  0.000000E+00
B054F08-09   74 -4.631390E-03  0.000000E+00
B054F08-09   75 -4.723300E-03  0.000000E+00
B054F08-09   76 -3.346670E-03  0.000000E+00
B054F08-09   77 -7.932110E-04  0.000000E+00
B054F08-09   78  2.244770E-03  0.000000E+00
B054F08-09   79  4.865160E-03  0.000000E+00
B054F08-09   80  6.205830E-03  0.000000E+00
B054F08-09   81  5.727300E-03  0.000000E+00
B054F08-09   82  3.408610E-03  0.000000E+00
B054F08-09   83 -1.998010E-04  0.000000E+00
B054F08-09   84 -4.091930E-03  0.000000E+00
B054F08-09   85 -7.075960E-03  0.000000E+00
B054F08-09   86 -8.127910E-03  0.000000E+00
B054F08-09   87 -6.728310E-03  0.000000E+00
B054F08-09   88 -3.071940E-03  0.000000E+00
B054F08-09   89  1.923090E-03  0.000000E+00
B054F08-09   90  6.827210E-03  0.000000E+00
B054F08-09   91  1.009100E-02  0.000000E+00
B054F08-09   92  1.051750E-02  0.000000E+00
B054F08-09   93  7.664370E-03  0.000000E+00
B054F08-09   94  2.065940E-03  0.000000E+00
B054F08-09   95 -4.832190E-03  0.000000E+00
B054F08-09   96 -1.101000E-02  0.000000E+00
B054F08-09   97 -1.443760E-02  0.000000E+00
B054F08-09   98 -1.369340E-02  0.000000E+00
B054F08-09   99 -8.474570E-03  0.000000E+00
B054F08-09  100  1.738270E-04  0.000000E+00
B054F08-09  101  1.000400E-02  0.000000E+00
B054F08-09  102  1.808500E-02  0.000000E+00
B054F08-09  103  2.159350E-02  0.000000E+00
B054F08-09  104  1.866640E-02  0.000000E+00
B054F08-09  105  9.100940E-03  0.000000E+00
B054F08-09  106 -5.328700E-03  0.000000E+00
B054F08-09  107 -2.105410E-02  0.000000E+00
B054F08-09  108 -3.339580E-02  0.000000E+00
B054F08-09  109 -3.762260E-02  0.000000E+00
B054F08-09  110 -3.013700E-02  0.000000E+00
B054F08-09  111 -9.497550E-03  0.000000E+00
B054F08-09  112  2.299310E-02  0.000000E+00
B054F08-09  113  6.330400E-02  0.000000E+00
B054F08-09  114  1.053400E-01  0.000000E+00
B054F08-09  115  1.421240E-01  0.000000E+00
B054F08-09  116  1.672260E-01  0.000000E+00
B054F08-09  117  1.761340E-01  0.000000E+00
B054F08-09  118  1.672260E-01  0.000000E+00
B054F08-09  119  1.421240E-01  0.000000E+00
B054F08-09  120  1.053400E-01  0.000000E+00
B054F08-09  121  6.330400E-02  0.000000E+00
B054F08-09  122  2.299310E-02  0.000000E+00
B054F08-09  123 -9.497550E-03  0.000000E+00
B054F08-09  124 -3.013700E-02  0.000000E+00
B054F08-09  125 -3.762260E-02  0.000000E+00
B054F08-09  126 -3.339580E-02  0.000000E+00
B054F08-09  127 -2.105410E-02  0.000000E+00
B054F08-09  128 -5.328700E-03  0.000000E+00
B054F08-09  129  9.100940E-03  0.000000E+00
B054F08-09  130  1.866640E-02  0.000000E+00
B054F08-09  131  2.159350E-02  0.000000E+00
B054F08-09  132  1.808500E-02  0.000000E+00
B054F08-09  133  1.000400E-02  0.000000E+00
B054F08-09  134  1.738270E-04  0.000000E+00
B054F08-09  135 -8.474570E-03  0.000000E+00
B054F08-09  136 -1.369340E-02  0.000000E+00
B054F08-09  137 -1.443760E-02  0.000000E+00
B054F08-09  138 -1.101000E-02  0.000000E+00
B054F08-09  139 -4.832190E-03  0.000000E+00
B054F08-09  140  2.065940E-03  0.000000E+00
B054F08-09  141  7.664370E-03  0.000000E+00
B054F08-09  142  1.051750E-02  0.000000E+00
B054F08-09  143  1.009100E-02  0.000000E+00
B054F08-09  144  6.827210E-03  0.000000E+00
B054F08-09  145  1.923090E-03  0.000000E+00
B054F08-09  146 -3.071940E-03  0.000000E+00
B054F08-09  147 -6.728310E-03  0.000000E+00
B054F08-09  148 -8.127910E-03  0.000000E+00
B054F08-09  149 -7.075960E-03  0.000000E+00
B054F08-09  150 -4.091930E-03  0.000000E+00
B054F08-09  151 -1.998010E-04  0.000000E+00
B054F08-09  152  3.408610E-03  0.000000E+00
B054F08-09  153  5.727300E-03  0.000000E+00
B054F08-09  154  6.205830E-03  0.000000E+00
B054F08-09  155  4.865160E-03  0.000000E+00
B054F08-09  156  2.244770E-03  0.000000E+00
B054F08-09  157 -7.932110E-04  0.000000E+00
B054F08-09  158 -3.346670E-03  0.000000E+00
B054F08-09  159 -4.723300E-03  0.000000E+00
B054F08-09  160 -4.631390E-03  0.000000E+00
B054F08-09  161 -3.220800E-03  0.000000E+00
B054F08-09  162 -1.009000E-03  0.000000E+00
B054F08-09  163  1.303700E-03  0.000000E+00
B054F08-09  164  3.045970E-03  0.000000E+00
B054F08-09  165  3.768250E-03  0.000000E+00
B054F08-09  166  3.351660E-03  0.000000E+00
B054F08-09  167  2.014000E-03  0.000000E+00
B054F08-09  168  2.157850E-04  0.000000E+00
B054F08-09  169 -1.487520E-03  0.000000E+00
B054F08-09  170 -2.617400E-03  0.000000E+00
B054F08-09  171 -2.903110E-03  0.000000E+00
B054F08-09  172 -2.335680E-03  0.000000E+00
B054F08-09  173 -1.153850E-03  0.000000E+00
B054F08-09  174  2.507500E-04  0.000000E+00
B054F08-09  175  1.458550E-03  0.000000E+00
B054F08-09  176  2.142870E-03  0.000000E+00
B054F08-09  177  2.156850E-03  0.000000E+00
B054F08-09  178  1.556450E-03  0.000000E+00
B054F08-09  179  5.714310E-04  0.000000E+00
B054F08-09  180 -4.845180E-04  0.000000E+00
B054F08-09  181 -1.303700E-03  0.000000E+00
B054F08-09  182 -1.677330E-03  0.000000E+00
B054F08-09  183 -1.540470E-03  0.000000E+00
B054F08-09  184 -9.830210E-04  0.000000E+00
B054F08-09  185 -2.017990E-04  0.000000E+00
B054F08-09  186  5.594430E-04  0.000000E+00
B054F08-09  187  1.087920E-03  0.000000E+00
B054F08-09  188  1.256750E-03  0.000000E+00
B054F08-09  189  1.056950E-03  0.000000E+00
B054F08-09  190  5.814210E-04  0.000000E+00
B054F08-09  191 -7.991640E-06  0.000000E+00
B054F08-09  192 -5.334690E-04  0.000000E+00
B054F08-09  193 -8.541500E-04  0.000000E+00
B054F08-09  194 -9.031010E-04  0.000000E+00
B054F08-09  195 -6.943090E-04  0.000000E+00
B054F08-09  196 -3.156860E-04  0.000000E+00
B054F08-09  197  1.088910E-04  0.000000E+00
B054F08-09  198  4.565460E-04  0.000000E+00
B054F08-09  199  6.383650E-04  0.000000E+00
B054F08-09  200  6.233790E-04  0.000000E+00
B054F08-09  201  4.385630E-04  0.000000E+00
B054F08-09  202  1.558450E-04  0.000000E+00
B054F08-09  203 -1.358640E-04  0.000000E+00
B054F08-09  204 -3.536480E-04  0.000000E+00
B054F08-09  205 -4.485540E-04  0.000000E+00
B054F08-09  206 -4.085930E-04  0.000000E+00
B054F08-09  207 -2.607400E-04  0.000000E+00
B054F08-09  208 -6.093910E-05  0.000000E+00
B054F08-09  209  1.308700E-04  0.000000E+00
B054F08-09  210  2.637370E-04  0.000000E+00
B054F08-09  211  3.106910E-04  0.000000E+00
B054F08-09  212  2.687330E-04  0.000000E+00
B054F08-09  213  1.638370E-04  0.000000E+00
B054F08-09  214  3.296680E-05  0.000000E+00
B054F08-09  215 -8.591420E-05  0.000000E+00
B054F08-09  216 -1.608400E-04  0.000000E+00
B054F08-09  217 -1.788220E-04  0.000000E+00
B054F08-09  218 -1.408600E-04  0.000000E+00
B054F08-09  219 -6.493520E-05  0.000000E+00
B054F08-09  220  2.497510E-05  0.000000E+00
B054F08-09  221  1.048950E-04  0.000000E+00
B054F08-09  222  1.578430E-04  0.000000E+00
B054F08-09  223  1.748260E-04  0.000000E+00
B054F08-09  224  1.588420E-04  0.000000E+00
B054F08-09  225  1.188810E-04  0.000000E+00
B054F08-09  226  6.793200E-05  0.000000E+00
B054F08-09  227  1.898100E-05  0.000000E+00
B054F08-09  228 -1.898100E-05  0.000000E+00
B054F08-09  229 -4.095890E-05  0.000000E+00
B054F08-09  230 -4.795220E-05  0.000000E+00
B054F08-09  231 -4.395610E-05  0.000000E+00
B054F08-09  232 -3.296680E-05  0.000000E+00
B054F08-09  233 -1.997980E-05  0.000000E+00
B054F08-09  234 -1.098890E-05  0.000000E+00
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation, NR001 ch LHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 15
B057F04     Input sample rate:                     5.000000E+00
B057F05     Decimation factor:                     5
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             2.340000E+01
B057F08     Correction applied (seconds):          2.340000E+01
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain, NR001 ch LHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 15
B058F04     Gain:                                  1.000000E+00
B058F05     Frequency of gain:                     5.000000E-02 HZ
B058F06     Number of calibrations:                0
#		
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity, NR001 ch LHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           6.291300E+05
B058F05     Frequency of sensitivity:              5.000000E-02 HZ
B058F06     Number of calibrations:                0
